One thing you can look at once you have done this if issues persist is whether the interface card is being detected – there should be an entry in the device manager. I went back a few steps and decided to start with a fresh hard drive and NT 4. The unit is in great shape it just did not come with the PC or software that were previously used to control it. Example auction I have no affiliation: Here are the relevant stats.
|Date Added:||10 April 2013|
|File Size:||58.66 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Home Chromatography Gas Chromatography.
Where could I download the driver for GPIB card?? – Discussion Forums – National Instruments
I have the ISA card that is the 50 I 823550a. Example auction I have no affiliation: Alas, the problems persist. Follow us on Twitter: Subscribe to our eNewsletter with daily, weekly or monthly updates: I had a german version of windows xp and it had english UK as a default, i changed it to english US and it worked.
I agree with the last 8350a. I would truly appreciate some guidance from the ChomGurus!!!
I had the same problem and after trying everything the problem was tbe lenguage setting on xp. Connection to Chemstation PC.
GPIB INTERFACES – Agilent Technologies – Agilent Technologies 82350B PCI GPIB Drivers Download
Here are the relevant stats. I believe you will get operational within a few days with help from here.
I got about as far this time but I feel a lot more confident in moving forward. I think you need to configure the card manually not Auto Addand type in “hp” small hp even though the card can be You mention keysight io suite. Separation Science offers free learning from the experts covering methods, applications, webinars, eSeminars, videos, tutorials for users of liquid chromatography, gas chromatography, mass spectrometry, 8235a preparation and related analytical techniques. I recall that the GPIB addresses and names set up in the IO software have to match that indicated in the chemstation manual else things don’t work.
Also you can have a look to http: The unit is in great shape it just did not come with the PC or software that were previously used to control it. No registered users and 8 guests.
Instrument Control (GPIB, Serial, VISA, IVI)
I’ve used plenty with A. Latest Blog Posts from Separation Science.
Mon Aug 30, Okay I just made some relevant progress. The only other clue I have to go off of is an error message on startup that is produced by these events.
Is there a PDF on the Chemstation disk that explains things.
I noticed in a few screenshots that people had the option to select the Board in the IOConfig application and I did not. Consumer Products Guy Posts: Fri Sep 02, 5: With both drivers I now can add the board correctly.
Further into the breach! Mon Sep 13, 8: It does it for the injector as well but let us just remove the injector from the equation right now and focus on the GC. I still get the same 82530a message when I open Instrument online: Thu Sep 14, In total there are 8 users online::